Verification on the spot. New lower pricing – Checkit out!
Test your pipettes SAFELY accurately & instantly… Checkit yourself!
Having a service enter your facility to recalibrate your pipettes just isn’t worth the risk – or the wait. Verify the accuracy of your pipettes the safe way – Checkit® yourself and leave your worries behind.
Why take chances with your sample analysis?
Evaluate the accuracy of your pipettes in seconds with our convenient Pipette Checkit® cartridges. Low-cost, fast, and easy to use, the Checkit can save money on calibration services while giving you the confidence that your pipettes are delivering accurate volumes. No more trying to read a tiny droplet’s weight on a high-precision balance.
Whenever you need a fast, quantitative assessment of a pipette’s accuracy, just pull out a Checkit. Available for various size pipettes, the Checkit checks your pipette’s accuracy at the bottom, middle and top of its range. Wouldn’t it be nice to know whether a pipette needs recalibration, or if lab personnel could use a brush-up on their pipetting technique? With the Checkit, a clear answer is seconds away.
Many pipette manufacturers recommend regular in-laboratory checking of pipettes in addition to good maintenance and scheduled calibration. The Checkit provides a valuable early-warning system for a pipette that may be drifting out of calibration.
Confidence in your experiments
No worrying! Do your experiments with confidence, knowing that your pipettes are delivering accurate volumes.
Instant, easy verification, anywhere
No waiting! Evaluate pipette accuracy on the spot in just seconds – no matter where you are. Why mess with a high-precision balance?
Checkit is fast and easy. And no worries about evaporation or air currents. Your measured drop is encased in a glass tube within seconds, so the values won’t change while you’re trying to read them.
Instant checks anytime
Keep Checkits in the lab for instant confirmation of pipette accuracy. Especially useful if your pipette has been dropped, cleaned, or borrowed. Don’t take a chance when you’re setting up an experiment with precious samples or expensive reagents.
Perform checks regularly to watch for performance drift – get early warning before there’s a problem.
Train students and technicians
Good pipetting technique is an important part of laboratory success. Use the Checkit as a training tool to verify proper technique. The Checkit requires no special training and is easy to use even for novice lab members.
Download the Pipette Checkit brochure.
“Checkit is very easy and quick to use with excellent accuracy. It’s design is modern, simple and self-explanatory. The lightweight Checkit will be a nice and cool widget that any pipette user world love to add into her/his shelf.” Zhongyuan Kan, University of Pennsylvania
“We were able to give them a try, checking the calibration versus a number of recently calibrated pipettes. The Checkits looked pretty accurate and were very easy to use.” Patrick Ingram, University of Wisconsin
- Simple, user friendly operation
- Easy to read markings indicate volumetric accuracy
- Measure accuracy at the top, middle and bottom of the pipetting range
- Available in 7 models to accommodate various pipette volumes
- ISO 8655 calibration standards indication
- Dye used is conveniently included right within the cartridge
- Minimal evaporation of dye prior to reading results
- Elegant design
- Made in the USA
3 Key Advantages
Wherever, whenever! Get instant, easy verification of a pipette’s accuracy.
2. More accurate than 5 digital balances for low volumes.
Be confident that your pipettes are delivering accurate volumes.
Because you never know where your pipette has been!
Brief Intro to the Checkit
Use the Checkit® – Because You Never Know Where Your Pipette Has Been
Checkit 2 uL, box of 5$65.50 Add to cart
Checkit 5 uL, box of 5$64.40 Add to cart
Checkit 10 uL, box of 5$64.30 Add to cart
Checkit 20 uL, box of 5$63.20 Add to cart
Checkit 50 uL, box of 5$62.95 Add to cart
Checkit 100 uL, box of 5$62.75 Add to cart
Checkit 200 uL, box of 5$62.55 Add to cart
We also have a family of Checkits for robots and multi-channel pipettes.
Using the Checkit
Each Pipette Checkit® has three precision glass capillaries matched to the top, middle, and low end of the range of your pipette. Pipet the three diagnostic volumes into the matching sample wells, flip the wells into measurement position, and take your readings. The pipetted volumes are compared to easy-to-read graduations that tell you instantly whether the pipette needs adjustment.
Step 1: Draw dye into the pipette tip.
Place the Checkit on a level surface. Grasp the pull tab on the reservoir seal and peel it off to expose the measurement fluid. Fit your pipette with an appropriate pipette tip (not a filter or gel-loading tip). Set your pipette volume. Then draw the dye (measurement fluid) from the reservoir into the tip.
Step 2: Dispense dye into a well.
Pipet the liquid into the first well of the Checkit.
Step 3: Flip up the well tab.
Use your finger tip to rotate the well to the vertical position. The droplet will automatically be drawn into the measurement capillary.
Step 4: Checkit!
Use the scale printed on the Checkit cartridge to determine the actual dispensed volume. Then repeat for the middle (50% of largest volume) and bottom (10% of largest volume) of the range of your pipette.
Note: Always wear personal protective equipment (PPE) including goggles.
InstructionsInstructions on how to use the Pipette Checkit in PDF format.
Calibration vs. verification
The Pipette Checkit® is a tool to identify potential volume delivery errors for pipettes. It is not a calibration tool. If your institution requires pipette calibration on a specific schedule, you should adhere to this schedule and use the Checkit as an extra verification step between scheduled calibrations.
Expected performance at different volumes
Piston pipettes are most accurate at or near their nominal volume (the maximum volume the pipette can deliver). When a pipette is used near the bottom of its range, the measured volumes will vary more, and are also more strongly affected by slight variations in pipetting style or environmental factors like humidity. The markings on the Checkit reflect this variation.
A note on environment: One advantage of the Checkit is that it can be used in the same room in which the experiment will be conducted. This can help identify variation in delivered volume due to pressure, humidity, or temperature differences between the calibration environment and the experimental environment. Because the Checkit directly measures delivered volume rather than relying on gravimetric analysis, there is also no need to use correction calculations that account for these factors.
Accuracy and precision
Most pipette manufacturers provide specifications for both the accuracy and precision of their pipettes. Briefly, accuracy describes how close the mean of delivered volumes is to the stated volume, and precision describes the distribution of the volumes around the mean. If you choose to compare your Checkit readings to the manufacturer’s specifications, remember to account for both accuracy and precision when determining what range is acceptable.
Note that ISO 8655 standards differ from the manufacturer’s standards. If your pipette has been recalibrated since purchase, you should check with your calibration service to determine which standards were used, and consider comparing your Checkit readings to that standard rather than to the manufacturer’s standard.
How rapidly should the well tabs be lifted up?
Lift the well tab at a reasonably quick, comfortable rate, about as quickly as you would turn off or turn on a light switch.
When should the well tabs be lifted up?
To minimize evaporation losses, lift each tab immediately after dispensing dye into its well(s).
How hard do I need to push on the well tab? How far?
A light touch is all that is needed to lift the tab, until you feel it stop moving. Do not push hard, or you will risk breaking the end of the capillary. Leaving the tab up will reduce evaporation, thereby providing you more time to compare the meniscus to the adjacent markings.
How do I avoid air bubbles?
First when dispensing into the well, if you do a blowout, lift the end of the pipette tip out of the pool of dye in the well.
How do I center the dye in the well of a Checkit Pro?
You can touch your pipette tip to the bottom of the well to help find the center. On the smaller volume wells, you can feel a divot at the bottom to guide the pipette tip. You can also use your pipette tip, carefully, to help guide dye to center after pipetting. If your hands are shaky, it helps to use two hands when pipetting or ask someone else for assistance.
How accurate is the Checkit?
The Checkit Pro is accurate to within 1%, aside from external influences such as evaporation. The Checkit Go is accurate to within 2%. We will likely be able to substantiate 1% accuracy once the product has enough history.
What is the cross-hatched rectangle on the Checkit Pro?
ISO 8655 standard covers the accuracy requirements of hand held, single channel, adjustable pipettes. The rectangle illustrates the acceptable range. Note that the width of the ISO rectangles does not factor in the 1% accuracy of the pipette Checkit®.
Can the Checkits be used to test someone’s pipetting technique?
Yes, one good application of the Checkits is testing the accuracy of a person’s pipetting technique.
Can the Checkits be re-used?
No, the Checkits are designed to be disposable.
Can I replace the glass capillaries?
No. To ensure accurate readings, the capillaries are precisely positioned to within .001 to .002 inches (25 to 50 micrometers). It would be near impossible to refill the Checkit with such accuracy.
In what order should I test my pipette at the high, middle, and low end of its range?
We suggest that you test the high end of the range first, then the middle, and finally the low end. For example, with a 20uL Checkit Pro, first test it for 20 uL, then for 10 uL, and finally for 2uL.
Can I use the Checkit outdoors?
For accurate readings, the Checkit must be on a level surface, near room temperature. It will not be affected by altitude, humidity, or small changes in temperature.
Is the Checkit’s performance effected by changes in temperature?
Temperature should not have a significant effect on the Checkit between 10 – 30°C and its readings change less than 0.07% with a 10°C change. The evaporation effects are minimal once the droplet is in the glass capillaries in the Checkit. Thus making it more accurate than a scale. While the density of the water changes (and so the observed mass for a given volume) between 4-40°C, the glass capillaries are not subject to much thermal expansion over this range. Therefore, the volume should stay very close to accurate between these temperatures. Since the Checkit measures pipette volume with volume, there is no conversion of units as with a scale (from mg -> mL). Given the volume on the Checkit is read in the same experimental environment of your pipette, it is a more accurate reflection of how your pipette performs. This is an advantage of the Checkit over classical methods.
How do I dispose of the dye?
The dye is safe and non-toxic. It can be disposed down a sink drain.
Checkits for Robots / Multi-Channel Pipettes
The new Checkit® Go! Now loading…
If the page does not load after a few seconds, please try clicking the selected tab above again.